The Dark-Line Hemacytometer is of one piece construction, ensuring long term durability and accuracy. An H-shaped moat forms two counting areas, or plateaus. A “V” slash at the loading side of each plateau facilitates charing and reduces the possibility of overflow into the moat. Each plateau features enhanced Neubauer rulings. The ruled surface is 0.1mm below the cover glass, limiting the volume of blood or fluid over a square mm at 0.1cu.mm and over each of 400 squares (within the central square mm) to 0.00025 cu. mm. Contact of the flat, polished cover glass surface with cover glass supports produces an exact volume of fluid over the counting area.
Counting chamber only
Counting chamber with 2 cover glass
Complete set – Chamber with two cover glasses, pipettes and case.
Rulings cover 9 square millimeters. Boundary lines of the Neubauer ruling are the center lines of the groups of three. (These are indicated in the illustration below.) The central square millimeter is ruled into 25 groups of 16 small squares, each group separated by triple lines, the middle one of which is the boundary. The ruled surface is 0.10mm below the cover glass, so that the volume over each of the 16 small squares is .00025 cubic mm.
The number of cells per cubic millimeter = Number of cells counted per square millimeter X dilution (if used) X 10
The number of cells per milliliter = Number of cells counted per square millimeter X dilution (if used) X 10,000
One (1) Milliliter = 1000 cubic millimeters (cu mm)
One (1) Microliter (ul) = One (1) cubic millimeter (cu mm)
To clean the counting chamber: After completing the count, remove the cover glass and clean the counting chamber with water or a mild cleaning solution (10% solution of bleach). Dry the counting chamber with a soft cloth or wipe, or rinse with acetone.
If you are dissatisfied with your purchase, let us help you with a replacement, return or repair. Items can be returned within 7 business days of receipt of your order without any restocking fee. Items must be in new condition with the original packaging.
Please contact us to receive an RMA number and steps on how to proceed with your return.